Jazzy® 600 ES SpecificationsFDA Class II Medical Device*
Jazzy 600 ES
|Weight Capacity||300 lbs.|
|Maximum Speed3||Up to 4.0 mph|
|Front Wheels||6" solid OMNI-Casters|
|Drive Wheels||14" foam filled|
|Rear Wheels||6" solid OMNI-Casters|
|Drivetrain||Two motor, mid-wheel drive|
|Braking System||Regenerative and electro-mechanical|
|Suspension Type||Active-Trac® (ATX) with Mid-wheel 6® Technology|
|High-Back Seat (max. size)||20"W x 20"D|
|High-Back Seat-to-Floor Range||21” – 24”|
|Synergy® Seating||20”W 20”D|
|Synergy® Seat-to-Floor Range||17.25” - 19.25"|
|Standard Electronics||50A, Dynamic Shark|
|Battery Size (2 required)||12 volt, NF-22|
|Battery Weight5||NF-22 at 37 lbs. each|
|Standard Battery Charger||5A, off-board|
|Per-Charge Range3,6||Up to 17 miles|
|Weight of Heaviest Piece4||124 lbs.|
|Standard Seat Weight||18" wide - 51.8 lbs.
20" wide - 54 lbs.
|Standard Seat Size||16"x16"-18"
18" x 18"-20"
20" x 18"-20"
|Medicare Code Coverage||Group 2: Standard (K0822/K0823)|
|Warranty||5-year limited on frame;
13-month limited on electronics;
13-month on drive motors;
13-month limited on batteries.
1 Due to manufacturing tolerances and continual product improvement, this specification can be subject to a variance of + or - 3 %. Dimensions listed are for power base only. Overall measurements will vary based on seating and accessory selections.
2 Without front riggings.
3 Varies with user weight, terrain, battery type, battery condition and tire pressure.
4 Does not include seat or battery weights.
5 Battery weight may vary based on manufacturer.
6 Tested in accordance with ANSI/RESNA, WC Vol. 2, Section 4 and ISO 7176-4 standards. Results derived from theoretical calculations based on battery specifications and drive system performance. Testing conducted at maximum weight capacities. This specification can be subject to a variance of +10%, -5%.
|The information contained herein is correct at the time of publication; we reserve the right to alter specifications without prior notice.|